[Federal Register Volume 73, Number 101 (Friday, May 23, 2008)]
[Notices]
[Pages 30169-30173]
From the Federal Register Online via the Government Publishing Office [www.gpo.gov]
[FR Doc No: E8-11666]


-----------------------------------------------------------------------

NUCLEAR REGULATORY COMMISSION


Report to Congress on Abnormal Occurrences, Fiscal Year 2007; 
Dissemination of Information

    Section 208 of the Energy Reorganization Act of 1974 (Pub. L. 93-
438) defines an abnormal occurrence (AO) as an unscheduled incident or 
event which the U.S. Nuclear Regulatory Commission (NRC) determines to 
be significant from the standpoint of public health or safety. The 
Federal Reports Elimination and Sunset Act of 1995 (Pub. L. 104-68) 
requires that AOs be reported to Congress annually. During Fiscal Year 
2007, eleven events that occurred at facilities licensed or otherwise 
regulated by the NRC and/or Agreement States were determined to be AOs. 
The report describes five events at NRC-licensed facilities. The first 
NRC-licensee event involved radiation exposure to an embryo/fetus. The 
other four NRC-licensee events were medical events, as defined in Title 
10, Part 35, of the Code of Federal Regulations (10 CFR part 35). All 
five NRC-licensee events occurred at medical institutions. The report 
also describes six events at Agreement State-licensed facilities. 
[Agreement States are those States that have entered into formal 
agreements with the NRC pursuant to Section 274 of the Atomic Energy 
Act (AEA) to regulate certain quantities of AEA licensed material at 
facilities located within their borders.] Currently, there are 34 
Agreement States. All six events that occurred at Agreement State-
licensed facilities were medical events, as defined in 10 CFR Part 35, 
and occurred at medical institutions. As required by Section 208, the 
discussion for each event includes the date and place, nature and 
probable consequences, the cause or causes, and the actions taken to 
prevent recurrence. Each event is also being described in NUREG-0090, 
Vol. 30, ``Report to Congress on Abnormal Occurrences: Fiscal Year 
2007.'' This report is available electronically at the NRC Web site 
http://www.nrc.gov/reading-rm/doc-collections/nuregs/staff/.

I. For All Licensees

A. Human Exposure to Radiation From Licensed Material

    During this reporting period, one event at an NRC-licensed and 
regulated facility was significant enough to be reported as an abnormal 
occurrence (AO).
NRC07-01 Human Exposure to Radiation at Washington University Medical 
Center in St. Louis, Missouri
    Date and Place--May 29, 2007, St. Louis, Missouri.
    Nature and Probable Consequences--Washington University Medical 
Center (the licensee) reported that cancer treatment to a 22 year old 
patient using iodine-131 resulted in a dose to an embryo/fetus. On May 
29, 2007, the treatment was conducted at Barnes Jewish Hospital, the 
affiliated teaching hospital of Washington University School of 
Medicine, using 4.64 GBq (126 mCi) of iodine-131. Prior to that

[[Page 30170]]

treatment, the patient saw her prescribing physician on May 22, 2007, 
for a related consultation. In addition, because hospital procedures 
require a pregnancy test within 1 week before the therapy is 
administered, the licensee conducted a pregnancy test on the patient on 
the same day. That test yielded a negative result and the patient was 
advised not to get pregnant prior to the treatment. Moreover, before 
treatment on May 29, 2007, the patient signed a statement that, to the 
best of her knowledge, she was not pregnant. However, on May 30, 2007, 
the patient performed a home pregnancy test, which yielded a positive 
result. Consequently, the licensee performed another pregnancy test the 
same day, and the results indicated that the patient had been pregnant 
for 4-5 weeks at the time of the iodine-131 administration. The patient 
and the referring physician were informed of this event. As an 
approximation for the dose equivalent received by the embryo/fetus, the 
licensee's staff calculated an annual total effective dose equivalent 
to the patient's uterus, which was estimated to be 250-340 mSv (25-34 
rem).
    The NRC-contracted medical consultant confirmed the licensee's dose 
estimate and determined that the most likely result would be delivery 
of a normal infant (with regard to thyroid function) because the 
iodine-131 was administered at such an early stage in the pregnancy; 
however, the risk of childhood cancer may be slightly increased. The 
possible effects of the event have been discussed with the patient.
    Cause(s)--The causes of this event were the false negative 
pregnancy test and the patient's lack of awareness that she might be 
pregnant.
Actions Taken To Prevent Recurrence
    Licensee--Because the causes of this event were beyond the 
licensee's control, the licensee determined that no corrective action 
was necessary to prevent recurrence.
    NRC--There were no violations identified by the NRC.

II. Commercial Nuclear Power Plant Licensees

    During this reporting period, no events at commercial nuclear power 
plants in the United States were significant enough to be reported as 
AOs.

III. Events at Facilities Other Than Nuclear Power Plants and All 
Transportation Events

C. Medical Licensees

    During this reporting period, four events at NRC-licensed or 
regulated facilities and six events at Agreement State-licensed 
facilities were significant enough to be reported as AOs.
NRC07-02 Medical Event at St. Luke's Hospital of Kansas City, Missouri
    Date and Place--October 23-26, 2006, Kansas City, Missouri.
    Nature and Probable Consequences--On October 27, 2006, St. Luke's 
Hospital of Kansas City (the licensee) notified the NRC of a medical 
event that occurred during a high dose-rate (HDR) remote afterloader, 
using a 144 GBq (3.9 Ci) iridium-192 source, brachytherapy procedure to 
treat breast cancer.
    The authorized user physician developed a written directive that 
prescribed 10 fractionated doses, to be administered to the patient's 
left breast using a balloon catheter technique, with each dose 
consisting of 3.4 Gy (340 rad), for a total dose of 34 Gy (3,400 rad). 
The first fractionated dose was administered to the patient on October 
23, 2006. On October 26, 2006, after the seventh fraction and prior to 
administering the eighth fraction to the patient, the chief physicist 
noted a discrepancy. The investigation into the discrepancy revealed 
that the catheter length entered into the treatment planning computer 
was 93.0 cm (36.6 in), rather than 95.0 cm (37.4 in). This error 
resulted in delivering an unplanned dose of 100 Gy (10,000 rad), 1.0 cm 
(0.4 in) from the treatment site and proximal from the balloon. The 
area proximal from the balloon would have received an intended dose of 
24.5 Gy (2,450 rad), had the treatment been delivered as prescribed by 
the authorized user physician. Moreover, because the prescribed dosage 
was not delivered to the correct location, the patient also received an 
under dosage to the distal side of the balloon. Specifically, the area 
intended to be treated received a dose in the range of 7 Gy to 10 Gy 
(700 rad to 1,000 rad) rather than the prescribed dosage of 34 Gy 
(3,400 rad). The patient and the referring physician were informed of 
this event. The authorized user physician did not expect any acute 
adverse medical effects to the patient as a result of the medical 
event, but indicated that surgery may be required in the future. The 
authorized user physician discontinued further treatments and plans to 
follow-up on the patient clinically.
    The NRC-contracted medical consultant expects some necrosis to 
fatty tissue in the overexposed region of the breast, within 2-4 
months.
    Cause(s)--The medical event was caused by the dosimetrist's failure 
to enter the correct catheter length in preparing the treatment plan 
parameters for the HDR brachytherapy treatment. In addition, the 
licensee's written procedures for implementing HDR treatment plans did 
not require verification of the treatment plan parameters to ensure 
that they were correct.
Actions Taken To Prevent Recurrence
    Licensee--The licensee initiated several immediate and long-term 
corrective actions to prevent recurrence. Specifically, those 
corrective actions included (1) Revising the procedures for HDR 
treatments to include verification of the catheter length and input to 
the treatment planning computer by both the medical physicist and the 
authorized user physician, (2) revising the treatment plan record to 
require that the authorized user physician and the medical physicist 
document the verification of the catheter length, and (3) conducting 
in-house training to ensure that staff are aware of the new procedural 
steps and to ensure that the prescribing authorized user physician and 
the medical physicist actively participate in the training.
    NRC--On March 14, 2007, the NRC issued a Notice of Violation 
related to this event.
NRC07-03 Medical Event at Hackley Hospital in Muskegon, Michigan
    Date and Place--January 8, 2007, Muskegon, Michigan.
    Nature and Probable Consequences--On January 8, 2007, Hackley 
Hospital (the licensee) notified the NRC of a medical event that 
occurred during a brachytherapy seed implant procedure to treat 
prostate cancer. The written directive prescribed a total dose of 120 
Gy (12,000 rad) to the patient's prostate using 41 iodine-125 seeds as 
permanent implants. According to the licensee, because the patient 
moved, only 7 of the prescribed 41 seeds were delivered to the prostate 
(the intended site), and the other 34 seeds were delivered to an 
unintended site located approximately 4 cm (1.6 in) inferior to the 
prostate. As a result, the prostate received a dose of approximately 13 
Gy (1,300 rad) rather than the prescribed dose of 120 Gy (12,000 rad) 
(~90% less than the prescribed dose). In addition, the unintended site 
received a dose of approximately 110 Gy (11,000 rad) and the patient's 
skin around the

[[Page 30171]]

unintended site received a dose of approximately 2.4 Gy (240 rad). The 
patient and the referring physician were informed of this event. The 
patient will require further treatment via external beam therapy in 
order to deliver the appropriate dose to the prostate.
    The NRC-contracted medical consultant agreed with the licensee's 
dose estimate and concluded that the risk for impotence is somewhat 
increased by the additional radiation dose to the unintended site as a 
result of the medical event. There may also be some risk of perineal 
tissue fibrosis and skin irritation, although the risk may not be 
significant enough to cause clinical concerns.
    Cause(s)--The licensee determined the root cause of the event was a 
failure to identify the patient's movement before continuing with the 
procedure. In addition, the NRC inspector determined that the licensee 
failed to develop adequate written procedures to provide high 
confidence that each brachytherapy administration was in accordance 
with the authorized user physician's written directive, as required by 
10 CFR 35.41. Specifically, the licensee's procedures did not include 
appropriate steps or guidance to ensure that radioactive sources were 
positioned in the patient in accordance with the written directive and 
treatment plan.
Actions Taken To Prevent Recurrence
    Licensee--The licensee's corrective actions to prevent recurrence 
included revising its written procedure to ensure that sources are 
positioned in the patient in accordance with the written directive, and 
ensuring that the staff implements those revisions.
    NRC--On June 20, 2007, the NRC issued a Notice of Violation related 
to this event.
* * * * *
NRC07-04 Medical Event at Kennedy Memorial Hospital in Turnersville, 
New Jersey
    Date and Place--October 25, 2006 (identified on December 8, 2006), 
Turnersville, New Jersey.
    Nature and Probable Consequences--Kennedy Memorial Hospital (the 
licensee) reported that a patient was prescribed a brachytherapy 
treatment of 145 Gy (14,500 rad) to the prostate gland for prostate 
cancer using 104 iodine-125 seeds, but instead received a dose of 145 
Gy (14,500 rad) to an unintended treatment site. The brachytherapy 
seeds were implanted under ultrasound guidance; however, a post-
treatment computed tomography scan showed that the implanted seeds were 
displaced inferior to the intended position, resulting in a dose of 
approximately 8 Gy (800 rad) delivered to the intended treatment site. 
The patient and the referring physician were informed of this event, 
and additional external beam radiation treatment was recommended.
    The NRC staff conducted a reactive onsite inspection on December 
12, 2006. The NRC-contracted medical consultant reviewed the case and 
agreed with the licensee's analysis and conclusions, stating that no 
significant adverse health effect to the patient is expected.
    Cause(s)--The medical event was caused by the licensee's failure to 
accurately identify the position of the prostate during the 
intraoperative ultrasound guidance procedure.
Actions Taken to Prevent Recurrence
    Licensee--The licensee revised its procedures, including the use of 
a contrast medium in the Foley catheter balloon to more clearly 
identify the bladder/prostate interface, and use of fluoroscopic 
imaging to confirm anatomical positioning and verify seed placement.
    NRC--There were no violations identified by the NRC.
NRC07-05 Medical Event at the University of Virginia at 
Charlottesville, Virginia
    Date and Place--February 2-4, 2007, Charlottesville, Virginia.
    Nature and Probable Consequences--University of Virginia at 
Charlottesville (the licensee) reported that a patient was prescribed a 
brachytherapy treatment of 30 Gy (3,000 rad) for treatment of cancer of 
the cervix using cesium-137 sources. Instead, the patient received 7.7 
Gy (770 rad) to the cervix and small volumes of the rectum and vaginal 
mucosa received doses greater than intended, ranging from 14.14 Gy to 
26.77 Gy (1,414 rad to 2,677 rad). Upon removal of the implant, the 
licensee discovered that the applicator had been loaded with a plastic 
radioactive source carrier insert that was approximately 4 cm (1.6 in) 
shorter than the intended 24 cm (9.5 in) insert, which caused the 
sources to be displaced from the intended position. The patient and the 
referring physician were informed of this event, and additional 
external beam radiation treatment was recommended.
    The NRC staff conducted a reactive onsite inspection on February 
12, 2007. The NRC-contracted medical consultant reviewed the case and 
agreed with the licensee's analysis and conclusions, stating that no 
significant adverse health effect to the patient is expected.
    Cause(s)--The medical event was caused by the licensee's failure to 
ensure that the insert was of the correct length before preloading the 
cesium-137 sources.
Actions Taken To Prevent Recurrence
    Licensee--The licensee revised its procedures, including measuring 
the length of the insert before loading the source, and limiting the 
supply of inserts in the source loading room to inserts of the length 
used for standard applicator treatments. The licensee also implemented 
additional staff training.
    NRC--On May 7, 2007, the NRC issued a Notice of Violation related 
to this event.
AS07-01 Medical Event at St. James Hospital and Health Center in 
Olympia Fields, Illinois
    Date and Place--November 29, 2006--December 20, 2006, Olympia 
Fields, Illinois.
    Nature and Probable Consequences--St. James Hospital and Health 
Center (the licensee) reported that a 75-year-old female patient 
received a dose to an unintended area of approximately 4 cm2 
(0.6 in2) of 20 Gy (2,000 rad), which was prescribed to 
supplement surgery and external radiation treatments for cancer of the 
uterus. The treatment used a high dose-rate (HDR) afterloader 
containing an iridium-192 source with an activity of 370 GBq (10 Ci). 
The source stopped 20 cm (7.9 in) short of the intended position; thus, 
the patient received none of the prescribed dose to the correct 
location. The patient and the referring physician were informed of this 
event. Over the next 4 weeks, the patient was treated for wet 
desquamation on both of her inner thighs, surrounded by a halo of 
erythema and the licensee continues to monitor the patient.
    Cause(s)--The medical event was caused by human error. The licensee 
entered an incorrect initial value into the treatment system, and the 
treatment plan was not reviewed by an authorized medical physicist 
during the subsequent three weekly treatment sessions. The error was 
identified during a chart audit before the next similar HDR treatment 
was planned.
Actions Taken To Prevent Recurrence
    Licensee--The licensee reviewed previous administrations to confirm 
that this event was an isolated incident. The licensee also developed 
new procedures requiring additional quality assurance steps, including 
the presence of a medical physicist during treatments. In addition, 
licensee personnel received additional training on the revised 
treatment procedures.
    State--The State conducted an investigation on January 8, 2007, and

[[Page 30172]]

issued a Notice of Violation. On March 8, 2007, the NRC-contracted 
medical consultant investigated the matter for the State and supported 
the licensee's conclusions. The State accepted the licensee's 
corrective actions on April 12, 2007.
AS07-02 Medical Event at Aroostook Medical Center of Presque Isle, 
Maine
    Date and Place--January 16, 2007, Presque Isle, Maine.
    Nature and Probable Consequences--Aroostook Medical Center (the 
licensee) reported that a patient received 148 MBq (4 mCi) of iodine-
131 for a whole body scan, instead of the prescribed 5.6 MBq (0.151 
mCi) for a thyroid uptake scan. On March 6, 2007 during a follow-up 
visit with an endocrinologist, it was recognized that the wrong scan 
was performed. The patient and the referring physician were informed of 
this event. Using the methodology in NUREG-CR-6345, ``Radiation Dose 
Estimates for Radiopharmaceuticals'', the licensee estimated that the 
administration of 148 MBq (4 mCi) resulted in a thyroid dose of 51.22 
Sv (153.7 rem). The licensee concluded that no significant adverse 
health effect to the patient is expected.
    Cause(s)--The medical event was caused by human error. The licensee 
failed to verify the prescribed dosage for a specific patient directly 
with the referring physician. In addition, a written directive was not 
completed for this procedure.
Actions Taken To Prevent Recurrence
    Licensee--Corrective actions taken by the licensee included 
revising procedures to improve communication with referring physicians, 
to allow the certified nuclear medicine technologist to speak directly 
with the referring physician or authorized user to confirm the type of 
test to be conducted. Also, written directives will be required for all 
administrations of iodine-131 in quantities greater than 1.11 MBq (30 
[mu]Ci).
    State--The State Radiation Control Program (RCP) performed an 
onsite investigation on May 24, 2007, and requested that the licensee 
take corrective actions to prevent recurrence. The RCP initially 
reviewed and accepted the licensee's proposed corrective actions during 
this investigation. The RCP issued a Notice of Violation on November 1, 
2007, and awaits the licensee's response.
AS07-03 Medical Event in New York
    Date and Place--March 7, 2007; (Licensee) New York.
    Nature and Probable Consequences--The licensee reported a 
brachytherapy medical event to the New York State Department of Health. 
The event involved a 31-year-old female patient with a history of 
vaginal cancer. The treatment involved the use of both cesium-137 and 
iridium-192 seeds. Each ribbon contained 8 seeds with an activity of 
1.855 milligram radium equivalent (118 MBq or 3.19 mCi). The patient 
was to be administered a total dose of 25 Gy (2,500 rad) via 
interstitial brachytherapy, to be delivered to the 0.5 Gy (50 rad) 
isodose line for a total treatment time of 50 hours.
    On March 6, 2007, the iridium-192 seeds and the cesium-137 seeds 
were placed into the patient. Late in the morning of March 7, 2007, the 
medical physicist performed a manual check of the treatment plan 
calculations, and discovered that the hand calculations indicated a 
significantly higher dose rate than was generated using the treatment 
planning software. The ensuing investigations revealed that the 
original treatment plan was in error. On March 7, 2007, after 27 hours 
of treatment, the seeds were removed from the patient.
    The patient received an estimated dose of 45.9 Gy (4,590 rad) to 
the treatment site, rather than the intended 25 Gy (2,500 rad). The 
rectal dose was 73 Gy (7,300 rad). The radiation oncologist disclosed 
that the patient is at risk for radiation cystitis, rectal proctitis, 
and more importantly, fistula formation between the rectum and the 
vagina. The patient and the referring physician were informed of this 
event. The patient will be monitored closely over the next year by both 
her gynecologic oncologist and the radiation oncologist. The patient is 
being treated with broad spectrum antibiotics, along with daily 
treatments in a hyperbaric oxygen chamber.
    Cause(s)--The primary cause was the use of an inappropriate Dose 
Rate Factor (DRF) in the treatment planning system. The value used 
corresponded to the DRF for air kerma, however, the seed strength 
entered was in milligram radium equivalent. Other causes and 
contributing factors included failure to check the treatment pre-plan 
before the seeds arrived although there was time to do so; failure to 
double-check the calculations either prior to the implant or shortly 
thereafter; use of a treatment planning system that underwent 
acceptance testing for cesium-137 and iodine-125, but not iridium-192; 
and lack of recent experience preparing a treatment plan using iridium-
192. Neither the physicist nor the radiation oncologist had prepared a 
treatment plan using iridium-192 in 6 years.
Actions Taken To Prevent Recurrence
    Licensee--The licensee changed its policy and procedures to require 
a check of calculations for any single-fraction brachytherapy 
treatment.
    State--The State plans to follow-up on the licensee's 
implementation of their new procedures during the next regularly 
scheduled inspection.
AS07-04 Medical Event at Memorial Mission Hospital of Asheville, North 
Carolina
    Date and Place--April 24, 2007, Asheville, North Carolina.
    Nature and Probable Consequences--Memorial Mission Hospital (the 
licensee) reported that a 19-year-old female patient was prescribed a 
dose of 1.24 MBq (33.4 [mu]Ci) of iodine-131 for a diagnostic scan to 
assess the health of her thyroid, however, she was administered a dose 
of 1235.8 MBq (33,400 [mu]Ci) on April 24, 2007. The licensee 
discovered the event when the patient returned the next day for her 
uptake scan. The patient was placed on a gamma camera and given a whole 
body scan. The spectrum was identified as iodine-131 and the uptake was 
concentrated in the patient's neck area, consistent with a thyroid 
uptake. As a result, the patient received a dose to the thyroid of 
approximately 287.3 Gy (28,728 rad). The patient and the referring 
physician were informed of this event.
    The patient received an ablative quantity of radioactive iodine and 
initially showed classic signs of thyroidoitis, including inflammation, 
swelling, pain, and difficulty swallowing. The patient has recently 
started taking a synthetic thyroid hormone.
    Cause(s)--The radiopharmacy provided the hospital an incorrect and 
mislabeled dose. The hospital failed to conduct a proper and accurate 
receipt survey on the package when it arrived in the hospital's nuclear 
medicine department. The nuclear medicine technologist, who performed 
the package receipt survey, failed to investigate the higher-than-
expected dose rate off the transport container to determine if anything 
unusual was present. The nuclear medicine technologist assigned to the 
patient failed to correctly and accurately assay the dose in the dose 
calibrator. A second nuclear medicine technologist who is supposed to 
perform a quality assurance (QA) check of the dose calibrator reading, 
taken by the nuclear medicine technologist assigned to the patient, 
failed to correctly and accurately read the dose calibrator. The 
nuclear

[[Page 30173]]

medicine technologist assigned to the patient failed to recognize that 
the number of counts obtained from the neck phantom used for the uptake 
scan baseline was unusually high for the quantity of radioactive 
material prescribed for the patient.
Actions Taken To Prevent Recurrence
    Licensee--The licensee ceased purchasing radiopharmaceuticals from 
the radiopharmacy that provided the incorrect and mislabeled dose. The 
licensee set aside a designated area for receiving shipments of 
radiopharmaceuticals and posted a list of expected dose rates per 
shipment (based upon contents of the shipment). The licensee redesigned 
the patient administration log to serve as a check list for QA, 
instituted procedural changes to include a one-meter survey of each 
diagnostic capsule while it is being counted in the neck phantom prior 
to administration, and implemented updated training to acquaint all 
nuclear medicine technologists with these new policies.
    State--The State radiation control agency conducted an 
investigation into this incident assisted by the State board of 
pharmacy. The licensee's actions to prevent recurrence will be 
inspected at their next regularly scheduled inspection.
AS07-05 Medical Event at University of Washington Harborview Gamma 
Knife of Seattle, Washington
    Date and Place--November 16, 2006, Seattle, Washington.
    Nature and Probable Consequences--University of Washington 
Harborview Gamma Knife (the licensee) reported that a patient who was 
prescribed to receive 18 Gy (1,800 rad) during a gamma knife treatment 
actually received 28 Gy (2,800 rad). The gamma knife contained 267.7 
TBq (7,236 Ci) of cobalt-60. The patient and the referring physician 
were informed of this event. The licensee concluded that no significant 
adverse health effect to the patient is expected.
    Cause(s)--The cause of the incident was determined to be human 
error. The prescribing physician prescribed 18 Gy (1,800 rad) and 
erroneously entered 28 Gy (2,800 rad). The physician entered the 
prescribed value into the computer treatment planning system, rather 
than having the medical physicist enter the value as is the usual 
procedure, resulting in a failure to follow an established procedure.
Actions Taken To Prevent Recurrence
    Licensee--Corrective actions taken by the licensee included a 
verification process to ensure that the prescribed treatment value is 
transferred from the treatment planning computer to the gamma knife 
computer prior to patient therapy. Also, a treatment plan signed by the 
treating oncologist, physicist, and neurosurgeon is now required. In 
addition, the treating oncologist and physicist will verify and initial 
the prescribed dose and isodose treatment parameters prior to patient 
therapy.
    State--The State reviewed the licensee's corrective actions and 
determined that the procedures were adequate to ensure that this type 
of event should not happen in the future.
AS07-06 Medical Event at Physician Reliance of Fort Worth, Texas
    Date and Place--August 22, 2007, Fort Worth, Texas.
    Nature and Probable Consequences--Physician Reliance (the licensee, 
dba Texas Oncology at Klabzuba) reported that a patient who was being 
treated for lung cancer, with a high dose-rate (HDR) afterloader and an 
iridium-192 source, received 2,500 cGy (2,500 rad) during the first 
fraction, instead of the prescribed dose of 500 cGy (500 rad). The 
patient was prescribed to receive five fractions with 500 cGy (500 rad) 
per fraction over five weeks. The incident was discovered following an 
independent physicist's review of the treatment plan. The patient and 
the referring physician were informed of this event. The patient's 
pulmonologist concluded that no significant adverse health effect to 
the patient is expected.
    Cause(s)--The incident occurred as a result of the incorrect 
isodose line being chosen and entered into the treatment planning 
system. The oncologist signed and approved the treatment plan and the 
radiation safety office performed a second calculation to check the 
treatment plan. The treatment planning system then normalized the 
calculations to the incorrect isodose line and delivered the resulting 
treatment. The calculation error was identified by an independent 
physicist prior to administration of the second fraction.
Actions Taken To Prevent Recurrence
    Licensee--The licensee's corrective action was to change their 
procedure to include a second check by a licensed medical physicist of 
all treatment plans.
    State--The State issued two violations related to this event: (1) A 
violation of 25 Texas Administrative Code (TAC) 289.256(p)(4)(A) and 
(B) was cited because the procedure as implemented was insufficient to 
ensure that a second check of the printed output of the treatment plan 
was performed to verify the accuracy of the planned treatment factors 
prior to treatment; and (2) a violation of 25 TAC 289.256(o)(1) and 
289.256(p)(1) was cited because the instructions of obtaining the 
authorized physician's signed and dated written directive for each 
therapeutic administration were not followed. In addition, the State 
reviewed the licensee's corrective action of changing their procedures 
to include a second check by a licensed medical physicist of all 
treatment plans.

    Dated at Rockville, Maryland, this 19th day of May 2008.

    For the U.S. Nuclear Regulatory Commission.
Annette L. Vietti-Cook,
Secretary of the Commission.
 [FR Doc. E8-11666 Filed 5-22-08; 8:45 am]
BILLING CODE 7590-01-P